This seller is the official distributor of the brand
Learn more about BiMedis TRUST
This seller had successful transactions
EMG equipment is used to assess muscular function, nerve conduction and reflexes. EP equipment measures and records electrical signals from brain in response to audio, visual, or somatosensory stimuli. This information is used together with other clinical information to assess brain sensory functions. Modern equipment is usually designed for both tasks.
1. Facilities should consider the following factors before making the purchase: reliability, safety, supplier service support, the initial cost of the unit and its accessories, and the cost of electrode replacement. Some additional factors to consider include computer software and combined usage as an EP unit. Systems that can be upgraded have an advantage.
2. To narrow down the list of devices to consider, facilities should decide on the intended applications. If the device will be used only for auditory EPs, devices without this capability can be eliminated, as can those that provide a comprehensive set of capabilities, including auditory, visual, and somatosensory EPs - since they would be more costly. However, if the device is to be used for a wide range of applications, only a comprehensive set of capabilities should be considered.
3. When purchasing any type of EP graphic recorder, buyers should consider some data management capabilities. A computer interface is highly recommended to transfer collected data to an information system. The unit should also be able to store data, preferably on a hard drive and removable media, and must have a waveform display capable of showing at least eight waveforms simultaneously. A printer to provide hard copies of collected information is desirable as well as an artifact-rejection option.
4. It is possible to expand EP unit capabilities to include electromyography and electro-oculography. This can be cost-effective because it often eliminates the need to purchase a dedicated system for those procedures.
5. EP users should be trained in using the systems and interpreting the results. Certification as a clinical neuro-physiologist is required, and the technician administering the tests should also receive training in patient preparation.
6. Depending on facility preference, the electric stimulator can be either sub dermal or surface. The auditory stimulator should be capable of producing broadband clicks and noise masking at the minimum. Insert earphones are also recommended, and the minimum signal produced should be 5 to 50 pulses per second at an intensity of 0 to 100 dB SPL. The visual stimulator should include a checkerboard capable of at least 1 to 50 reversals per second.